- Forum: Philips iE33 Echocardiography Questions and Answers
- Last Post: September 30th, 2015
- 1 Reply
- Post a Reply
|Doppler Modes||Color Doppler and CW/PW Doppler|
|Additional Specifications||The following transducers are Philips IE33 probes: Philips S5-1, Philips S8-3, Philips S12-4, Philips X3-1, Philips X7-2, Philips X7-2t, Philips S7-3t, Philips C5-1, Philips C5-2, Philips C8-5, Philips C9-4, Philips L11-3, Philips L9-3, Philips L15-7io, Philips D2cwc, Philips D5cwc, Philips D2tcd, Philips S7-2, Philips S8-3t, and Philips TY-306|
Note: We appreciate the support of individuals and companies making their documents and resources available to our community. If you find a document or resource that does not belong please let us know by emailing firstname.lastname@example.org
Recent Question ActivityAsk a Question
The CW has excessive artifact-as does the pedoff probeprobe change with another-problem repaims with unit-not probe.Also screen goes grey and turning off and on several times is the only way to get the CW to quit.Am I looking at a NAIM board or a...
sir i am getting 044 error
Keep your instrument running smooth
There are no threads in Philips iE33 Echocardiography Preventative Maintenance at this time.
Add Preventative Maintenance Post
There are no threads in Philips iE33 Echocardiography Reviews at this time.
Add a Review
Ultrasound Tech Expert: Component Failures Create Additional Issues...
X7-2t Transducer Available Now!...
The Philips iU22 and iE33 ultrasound systems are excellent general imaging, vascular and cardiology units that have gained a very large share of the...
See it in action
Press Releases and Articles