- Forum: GE Healthcare Precision 500 D Questions and Answers
- Last Post: November 21st, 2013
- 1 Reply
- Post a Reply
|Maximum Focal Spot||73.5 in|
|Motorized Tilting/Pivot Rotation||± 90|
|X-ray Generator Power Rating||80 Kw|
Recent Question ActivityAsk a Question
I have forgot whether I need G.E.'s service level software to do a full cal on this system? It's been a while. Anyone?
Is anyone interested in G.E. Precision or Revolution training at there site? I am offering a 3 day crash course on many G.E. X-ray systems for experienced and newbie x-ray system service technicians. Contact me if interested! firstname.lastname@example.org
Keep your instrument running smooth
There are no threads in GE Healthcare Precision 500 D Preventative Maintenance at this time.
Add Preventative Maintenance Post
There are no threads in GE Healthcare Precision 500 D Reviews at this time.
Add a Review
Press Releases and Articles