Class 1 Recall: Physio-Control Inc. Recalls Defibrillation Electrodes Due to Incorrect Placement Instructions for Infants Depicted on Artwork · a month ago by Other Author
Defibrillation electrodes used on the LIFEPAK EXPRESS® AED, LIFEPAK CR® Plus AED, LIFEPAK® 1000 defibrillator, or LIFEPAK 500 Biphasic AED with a pink connector.
Model/Item Numbers: 11101-000016 and 11101-000017
Lot codes: 713609, 717912, 713904, 718033, 715008, 719323, 45932237, 46042286, 45979590, 46050960, 45979954, 46052545, 46007867, 46061770, 46023185, 46063054, 46023823, 46078012
Manufacturing Dates: April 27, 2017 to August 10, 2017
Distribution Dates: May 30, 2017 to September 4, 2017
Devices Recalled in the U.S.: 7,973 units nationwide
Automatic external defibrillators (AEDs) are used to deliver lifesaving electrical shocks to people with sudden cardiac arrest, a medical condition in which the heart suddenly and unexpectedly stops beating. Defibrillation electrodes are connected to the AED to help the device analyze a patient's heart rhythm and deliver an electrical shock to restore normal heart rhythm when needed. The primary users of AEDs are first responders and hospital health care providers.
Physio-Control Inc. Infant/Child Defibrillation Electrodes
Reason for Recall
Physio-Control Inc. is recalling infant/child defibrillation electrodes because the artwork on the pads within the packaging shows incorrect placement instructions for infants. There is no issue with the performance or function of the defibrillation electrodes. However, incorrect placement of the electrodes on an infant may result in failure to deliver an effective shock to an infant in cardiac arrest. A delay in therapy could result in serious injury and/or death.
Who May be Affected
Infant patients who may need these defibrillation electrodes to restore normal heart rhythm.
Health care providers and first responders using these defibrillation electrodes on infant patients.
What to Do
On October 27, 2017, Physio-Control Inc. issued a Voluntary Field Action to provide customers with correct electrode placement instructions to be included with the AEDs until they receive their corrected defibrillation electrodes. As an alternative, if customers decide not to use the affected defibrillation electrodes and they do not have a spare set of infant/child defibrillation electrodes, customers may consider the use of adult defibrillation electrodes based on American Heart Association and European Resuscitation Council 2015 Guidelines until they receive their replacement set of infant/child defibrillation electrodes.
Customers with questions regarding this notification, please contact Physio-Control by calling 1-866-231-1220, Monday – Friday, between 6:00 am to 4:00 pm (Pacific time), or by email to firstname.lastname@example.org or fax to 1-866-448-9567.
Date Recall Initiated
October 13, 2017
How do I report a problem?
Health care professionals and consumers may report adverse reactions or quality problems they experienced using these devices to MedWatch: The FDA Safety Information and Adverse Event Reporting Program either online, by regular mail or by FAX to 1-800-FDA-0178.