Have PM Tips to Share? Send Them In!
Submit your preventative maintenance and other industry tips in today!
Mon Apr 12 2021
At MedWrench, our mission is to be the go to resource for the medical equipment industry. This starts with sharing tips with each other! Fill out the form below to share some industry tips and advice!
"Best practice for the use of any pressure measurement device is to start-up/zero the device with NOTHING connected. This eliminates the possibility of the device self-calibrating to a false (not quite) 'zero'." - Jim Munro
"They told me that the device should be turned on with your “T-Tubes”, which attach between the SimCube and your BP test cuff, NOT connected to the NIBP hose or even the SimCube itself, until you see the zeroes. Sometimes, when turning on the device with the hose connected to the SimCube, but not the NIBP hose, it can be hard to clear those lines. In that case, take your thumb, put it over the connection to the test cuff and put it over the opening a couple of times (See Figure 2). If that doesn’t work, try powering it on without the t-hose connected to the device." - Deanna Wilkinson
"Use your Computerize Medical Maintenance Software (CMMS). Today’s CMMS have many automated capabilities that will allow automated delivery of the work order to the originator and their leader." - Tony Cody