The AMSCO 3052 Single-Chamber Washer/Disinfector is designed for use in the cleaning and intermediate-level disinfection of soiled reusable utensils, trays, bedpans and urinals.
The AMSCO 3052 Single-Chamber Washer/Disinfector is designed for use in the cleaning and intermediate-level disinfection of soiled reusable utensils, trays, bedpans and urinals. It can also be used for rubber and plastic goods, simple hard-surfaced rigid surgical instruments (such as forceps and clamps) and other similar and related articles found in healthcare facilities.
FORUMSView All (2)
4 years ago
|16. Chemical Injected Volume Too Low|
During Instrument cycle on Wash 2, chemical used was Detergent. While injecting, washer alarmed "16. Chemical Injected Volume Too Low." The alarm would pop up in the middle of the day, which made me believe that it wasn't a check valve issue next to float sensor for the chemicals. I replaced all hoses for the Knight peristaltic pumps and added silicone grease. I disassembled all valves and flow sensors to check its condition (all seemed in good condition), so I was thinking that it would be a faulty intermittent flow sensor on detergent valve. I also calibrated volume delivery accuracy for all pumps, and yet it failed with the same alarm. I recently replaced detergent valve with an upgraded kit (Chemical Valve Installation PN: 10094478). It ran fine for a week and now the error came back. Has anyone ever encountered the same issue? Any suggestion would help!Reply
4 years ago
|In Search of Service Manual|
Does anyone have a service manual they can email me please! My email is email@example.com, Thanks!!!Reply