Nothing compromised the imaginative thinking that went into designing the LOGIQ E9 premium ultrasound system.
LOGIQ E9 has advanced Volume Navigation tools that represent another milestone in the GE Healthcare volume ultrasound experience. This sophisticated technology empowers users with the agility to do more with an ultrasound system than ever before in superficial and abdominal applications.
Volume Navigation helps to increase diagnostic confidence by combining multi-modality imaging with any ultrasound mode, including Contrast Enhanced Ultrasound and CFM. LOGIQ E9 knows the next step of a scan and helps users get there like no other ultrasound system can. It’s all part of the technologically advanced Scan Assistant, customizable scanning protocol. Designed by users, Scan Assistant enables users to take more control over the important elements of an exam by doing the little things for users. That’s because it automatically inserts comments, initiates and autocompletes required measurements, and much more. Scan Assistant does things that an ultrasound system has never done before, including automatically steering color Doppler and setting up imaging controls and modes for users. Now users can truly concentrate less on keystrokes and more on patient care.
FORUMSView All (55)
a month ago
|System error, please restart|
Hello. I have a GE logiq E9 ultrasound machine which displays "system error" and please restart the system. I reinstalled the "software application". a few moments after the error comes back. However I don't have the basic software but only the "R2.0.4 sofware application". However I have another CD with the full backup. My questions are as follows: 1) What is this error? 2) How do I restore the system from my "full back up" CD? If you can tell me the procedure. 3) Which version of "base system software" is compatible with "application software R2.0.4"?Reply
a year ago
|Clear Dicom Worlist|
I have run the utility to remove patient data and that seemed to work. The patient file and imag file are now empty. However I still have a handful of patients in the DICOM worklist that I cannot seem to get rid off. This system is not connected to a network. Any ideas would be greatly appreciated.Reply
a year ago
|PW knob not responding properly|
I replaced the GTX board and now the PW knob is not responding well, it works for few seconds and stops. Kindly advise on what to do, will offset calibration correct this problem? And how to I do that?Reply
SERVICE COMPANIESView All Ultrasound Systems Companies
GE Logiq E9 - DICOM Setup...
GE Logiq E9 - Creating a Backup...
GE Logiq E9 - Software Installation...
|Doppler Modes||3-D, Color, Continuous wave (CW), Pulsed wave (PW), Power Doppler|
|Image Display||19 in Color Flat Panel|