The ProSound SSD-4000 utilizes the most advanced acoustic technologies available today, and its multidisciplinary technology architecture enables it to offer great versatility and flexibility over a wide range of clinical applications.
With its new-generation, front-end technology including a 12-bit A/D converter, the ProSound SSD-4000 offers superior contrast resolution—especially when compared to 10-bit systems.
An intelligent digital beam former supports PixelFocus™, providing precision spatial resolution down to the pixel level for improved visualization of clinical detail.
The SSD-4000 utilizes newly developed Wide-band Super High-Density (W-SHD) transducers optimized for digital imaging.
The system also employs a Pure Harmonic Detection (PureHD) feature to create exceptionally clear images—artifact free and with a higher signal-to-noise ratio. This is especially useful when scanning traditionally difficult-to-scan patients (e.g., the aged or the obese).
The PureHD technology enables distortion-free fundamental frequency transmission as well, which makes harmonic images much clearer. This maximizes the affect of both the Tissue and the Contrast harmonic echo by providing excellent contrast and spatial resolution.
18 days ago
|Error message about Tx voltage|
Hallo, i have e problem about SSD 4000; when the equipment start compare on the Monitor display this error message: "Power for ulrasound transmission was shut off as the system detected abnormal drive voltage. Please reboot the system." If i don't connect any probe the display image is Ok, when i connect any probe a error message compare! I make a Flash Rom update whit Foppy disk but nothing change. Have you a suggestion, please. Thank you.Reply
a month ago
Aloka ssd 4000 needs a service manual, it starts but doesnt initialize all hardware. the cMOS battery was down adn after replacing, we couldnt get it to fuilly work againReply
6 months ago
|Operating System & Installation|
Hello, I am installing this machine for the first time, but i don't understand the operating system to install and the entire process. Could you please guide me on how to do it? I also need the product manual.Reply
|Doppler Modes||Color, Continuous wave (CW), Pulsed wave (PW), High pulse repetition frequency (HPRF)|
|Image Display||15 in Color CRT|
Number of Channels:256