DESCRIPTIONCompact, tabletop digitizer for computed radiography
The CR 30-X digitizer for computed radiography (CR) is a compact, tabletop digitizer for decentralized CR environments. It combines high image quality with high throughput. Easy to install, use and maintain, it provides seamless integration, from X-ray exposure to print or softcopy.
Using dedicated cassettes that are inserted horizontally, the CR 30-X requires minimal space and easily integrates in vans, trucks and other mobile facilities, or other places where space is an issue.
CR 30-X offers an ideal solution for private and decentralized CR environments, including decentralized hospital departments, orthopaedics, chiropractics, military and veterinary applications.
FORUMSView All (196)
6 days ago
|Mammo film printing|
Hi My "Drystar 5503 " doesn't print mammo exams in small film format even 8*10 inch films are in the input tray. its gives nothing and show nothing in NX workstaion interface. i check the printer input tray settings it gives me that films are in and in right format. i had print test image from the tray that contain Mammo films and it works fine. then i had change printer configuration from NX workstation i gives Mammo cassette 8*10 format. then its starts print out small images in 14*17inch films please how can i solve this problem Reply
19 days ago
|NX viewer Ref.MESS243|
Hi My NX viewer starts to give this message every time i entered CR cassette and select new exam Message:" unable to identify the cassette [Ref.MESS243] if a cassette not dedicated to mammography is inserted: Please identify a "non mammo" image or: please identify an image dedicated to mammography" i select every time the type of exam ex: mammo or abdomen but its doesn't work when i click to ID button it show that message! please how can i solve this problemReply
a month ago
|CR gives error 14860 at startup|
Hi, The CR gives error 14860 at start-up and in service mode it says polygon doesn't reach the speed, indicates polygon or power supply fault, we changed the optical module and the power board but still same error, any help?Reply
DOCUMENTS / MANUALSView All
High image quality and throughput The CR 30-X scans at a spatial resolution of 10 pixel/mm, then processes the image using MUSICA² - Agfa HealthCare's patented, market-leading intelligent image processing software. By scanning the phosphor plate at the full width of 43 cm, the CR 30-X provides both high quality and high throughput.
The CR 30-X supports all common cassette formats.
Low total cost of ownership Installing the CR 30-X digitizer can be achieved in a single day. Its special LED technology in the erasure unit keeps the heat output and the power requirements lower, so that standard electrical outlets are sufficient. The lack of preliminary electrical work means the set-up cost is lower. And with its modular, component-based design, the CR 30-X digitizer offers faster, easier and more cost-effective maintenance.
Facilitates convenient and fast workflow The CR 30-X works in conjunction with Agfa HealthCare's NX - the CR's image identification and quality control workstation designed to enable users to complete effortlessly, and with minimal training, a complete digital X-ray exposure. In addition, the CR 30-X uses MUSICA intelligent image processing software to automatically optimize the final image quality without the need for human intervention.
Designed to enable a seamlessly integrated digital workflow, the CR 30-X uses DICOM protocols for all steps in the digital workflow. The CR 30-X can be used with following Agfa HealthCare products and services:
- CR MD 4.0T
- MUSICA 2
- DRYSTAR 5300
- DRYSTAR 5302
|CR-System Configuration||Single-plate table unit|
|CR-Workstation Configuration||Included with each image reader|
Throughput (@ 14x17, std resolution):60 plates per hour