DESCRIPTIONHigh Pressure/High Flow Oxygen Concentrator
NewLife Intensity combines high pressure with high flow to create the most power-efficient 8 liter per minute (LPM) oxygen concentrator, operating at only 410 watts for reduced power consumption. With an outlet pressure approximately three times higher than most other concentrators, this unit is designed to power respiratory accessories including large volume jet nebulizers, venti-masks, medication nebulizers, and long oxygen tubing runs.
The NewLife Intensity 8 LPM oxygen concentrator simplifies a tracheostomy set-up by eliminating the need for a separate external compressor. Administer humidified oxygen to a trach patient by connecting a jet neb bottle directly to Intensity’s O2 outlet. The NewLife Intensity dual flow unit can provide additional power savings by delivering oxygen to two patients simultaneously up to a combined total flow of 8 LPM with one unit.
FORUMSView All (2)
13 days ago
Sir my new life oxygen concentrator(10 litres) starts beeping just after 15-20 mins of start of the machine and it si continious. I switch off and again switch it on and again beep is gone but after 15 mins again beep is there. What would be the cause. Please helpReply
4 years ago
|AIRSEP INTENSITY DUAL FLOW 8LPM|
Sir, I have a Airsep Intensity dual flow 8 LPM.Upon turning on,
Problem Found: Compressor relief valve releases and producing Alarm
Could you help meReply
DOCUMENTS / MANUALSView All
Oxygen Concentration: 2–7 lpm: 92% ±3% 8 lpm: 90% ±3%
Oxygen Outlet Pressure: 20 psig (138 kPa)
Power Consumption: 410 Watts