A pleasant atmosphere, comfort, and short examination times are becoming increasingly more important in the radiology department. This is where our MULTIX systems really shine.
3 months ago
Siemens - APID 080 - Error 119/080Reply
9 months ago
|Multix Collimator detent indicator issue|
I have a Multix Top with a transverse detent " Amber Indicator" that does not come one when in position. I have talked with Siemens, went through the software calibrations. I have compared these calibrations with another room and these are the same.
This indicator would be illiuminated when using the chest stand, This indicates proper alignment. The longitudal indicator works as it should. During power on the indicators are lit so no issues with the LED.
The hard copy of the service manual has a lot of voids to try to trace what is suppose to happen and when.
Thanks in advance for advice.
a year ago
|x ray tube noisy|
Our anode tube make noise in vertical position. If we change to the wall position the roar desappears.Any hints?Reply
Siemens Announces FDA Clearance of ACUSON SC2000 Prime Ultrasound System
Digital imaging solution offers state-of-the-art X-ray technology for smaller budgets.
Study findings suggests potential use of [18F]-T808 as PET imaging agent for tau protein pathologies associated with Alzheimer’s disease