DESCRIPTIONA 64-slice CT scanner with Essence technology that delivers value, dependability, and performance.
The Philips Brilliance CT 64-slice is one of the most advanced CT systems in the market. This system can expand clinical boundaries in cardiac, pulmonary, trauma, and pediatric imaging.
This system is built on innovative Essence technology that delivers image quality, dose efficiency and rapid reconstruction times.
The Brilliance CT 64 channel configuration package embodies completely the philosophies of sense and simplicity applied to more effective patient care. With a range of new and targeted advancements, including low-dose cardiac imaging, 3D volume software and expanded portal capabilities, this system takes the raw capabilities of volume imaging and makes them work harder for patients by making the physician’s work easier. This is the system for those who want the latest CT can offer, and realize productivity is an integral component of results.
FORUMSView All (16)
3 days ago
|COUCH´S VERTICAL MOVEMENT FAULT|
The system won´t move the couch in the vertical direction. If I press the buttom switch on the gantry, the leds goes off for a while, but the couch doesn´t move. However, when I select the service switch, the couch can be move up and down. I already try to flash the CPM, check interlocks and voltages.Reply
7 months ago
|CARDIAC SCAN ERROR "Gantry acquisition system cannot comply"|
WHEN I AM DOING THE CARDIAC ACCORDING TO THE BOLUS TRACK ,SCAN WILL START BY SELF AND IT WILL STOP IMMEDIATELY .iTS HAPPENING ONLY IN THE CARDIAC SCAN ,ALL OTHER BOLUS TRACK SCANS ARE GOING WELL.aLSO FOLLOWING ERROR IS COMING UP "Gantry acquisition system cannot comply"Reply
a year ago
|EBW Software Installation manuel|
Does anybody has Phiips Brilliance EBW software installation manuel?Thanks in advanceReply
SERVICE COMPANIESView All CT Scanner Companies
|Gantry (Max. Clearance)||70 cm|
|Maximum Scan Range||175 cm|
|Patient Weight Capacity||450 lbs|