The ACUSON Sequoia™ echocardiography system provides the most advanced imaging and clinical applications across your ultrasound specialties. The Sequoia system provides superior imaging for every patient, with Native Patient Specific Imaging.
Native TEQ™ ultrasound technology ensures consistent image quality while reducing operator workload.
The ergonomic articulating arm and flat panel display reduces glare in all working environments, enhances operator comfort, and folds down for easier system transport.
The Sequoia system’s legendary upgradeability and longevity ensures that your investment is protected for years to come.
a month ago
|Acuson 512 setup.|
Is there anyone familiar with setting the 512 with a PACs network. I need to setup a way for my client to make CDs to give to his Patients. Or is there away to make CDs directly for the Acuson 512 ultrasoundReply
3 months ago
Do anybody knows where is position of RAM at Acuson Sequoia C512?Reply
4 months ago
|acuson c512 cw and pw error|
when i am doing cardiac scan if i press CW or PW button for graphs, that time machine got stuck (hang), to start the machine we need to reset and restart. which board is related to CW and PW mode?Reply
3 years ago
|Biomed tip for Networking Sequioa Ultrasound system|
Biomed Tip for Sequoia Ultrasound System Networking The Sequoia ultrasound system is very particular when it comes to Network / Dicom setup and editing. Sequoia networking setup is a more advanced level of interaction with the Sequoia, if you are not familiar with the service tools this maybe a task you ask in which you ask for help. Provided is a list of guidelines that will help keep you out of trouble when setting up or editing a Sequoia network. These guidelines are the same for creating a new network or editing a network. We call them guidelines because with the Sequoia and networking it never seems that there are hard fast rules. It is realized that there is more than one method to accomplish these edits to the network configuration. However, 15 years of Sequoia experience provides the safest and best guidelines for success. Sequoia Networking Guidelines: Always create a SYSTEM and CONFIG backup before any Network creation or editing. (SYSTEM contains all Sequoia system specific information; CONFIG contains the Network setup information.) Create or edit a network configuration by booting to imaging and selecting Service UI from the setup button. Never boot to service to create or modify a network. When creating a new network configuration, called Network Host List, start fresh and do not edit the “service” configuration. Always enter a router but do not name it “router”, name it something other, “gateway” works well. Never edit the Sequoia 127.0.0.1 loopback address, this should be preset in every network and the Sequoia should not let you edit it. It should always be a “Network Host”. Never edit an existing device, always delete it and re-enter the device as a whole. Never select the “edit in all network” or “delete from all network” button. The Sequoia itself is always a “Hostname”. When changing Network speed it is often that you must turn the unit off not just let it reboot. Sometimes a reboot alone can cause unpredicted results. When a Perspective PC is involved you cannot edit the Sequoia address or the Perspective PC address from the currently active Network. You must first switch to a different network scheme often one exists called “Standalone”. Failing to do this properly will result in the Perspective PC not functioning with the Sequoia. If you have problems use the backup disk to restore your settings. When all done create new SYSTEM and CONFIG backups. Reply
-Ted Lucidi, CBET
6 years ago
|Routine cleaning of connector ports|
Why? The connector ports on the Acuson Sequoia are a place where dust and lint can and will collect. In order to cool the scanner, air flow is normally routed through the scanner’s main filters. As the filters become more restricted, air will take the path of least resistance into the scanner. The design, of the micro-pinless connectors on the Sequoia, makes the collection of dust and lint extremely problematic. Dust and lint become trapped on the small, pointed pins on the connector’s surface. This build-up of debris can and will, over-time, cause a variety of intermittent issues on all types of probes. • Probe ID failures • Electro-magnetic interference (EMI) susceptibility • Calibration failures • Dropout and image quality degradation How? Ports should be cleaned with the scanner power OFF. Use a soft-bristled brush to gently wipe the connector plate. Compressed air or a vacuum is recommended to aide in the elimination of the dust. When? This process should be integrated into a comprehensive PM and filter cleaning program. Many of our customers have experienced great success in minimizing intermittent issues with routine filter and connector cleanings, either: • Bi-weekly • Monthly • Or based on filter monitoringReply
|Doppler Modes||Color, Continuous wave (CW), Pulsed wave (PW)|
|Image Display||17 in Color Flat Panel|
Siemens today announced the addition of a new, compact ultrasound system to its Healthcare sector portfolio that enables physicians to obtain fast and reliable diagnoses under difficult conditions.