DESCRIPTIONThe Burdick Elite II ECG / EKG machine provides ECG reports quickly and easily from an affordable; compact package.
The Burdick Elite II ECG / EKG machine provides ECG reports quickly and easily from an affordable; compact package. Choose the Burdick Elite II for accurate analysis and simple operation in a portable unit. Perfect for insurance examiners.
FORUMSView All (4)
2 years ago
how to use ekg Reply
6 years ago
|KIM MEDICAL DIRECTOR|
I HAVE A BURDICK ELITE EKG. HOW DO I GET IT TO PRINT IN ORDER
i,ii,ii,ARL AVL AVF, V1-V6
NEED TO KNOW ASAPReply
7 years ago
|Manual for Burdick Elite|
I don't have a manual for a Burdick Elite.
I'm trying to determine how to change the format of the EKG print out to the tradition I, II, III, AVR... print out.
SERVICE COMPANIESView All Electrocardiograph (EKG / ECG) Companies
- Easy, one-touch operation.
- Portable; weighs just 9 pounds.
- Battery or AC operation.
- Stores up to 30 ECGs in solid-state memory.
- Prints in single-channel or 12 lead format for easy mounting.
- Simultaneous acquisition and analysis of all 12 leads.
- Uses the Glasgow Royal Infirmary (GRI) Interpretive Algorithm for documented accuracy.
Display: 2 x 20 character LCD
Keyboard: Numeric entry: hardkey control Input/output:
Serial: RS-232 (9-pin D);
Signal options: 1 mV input - 1 V output +/_ 1-%
Power requirements: AC operation: 115/230 V AC +/- 10%, 50/60 Hz;
Battery operation: 12.5 V NiCad rechargeable battery pack.
Printout: Printout device: 48 mm thermal dot array;
Paper dimensions: 50 mm wide, 45 mm grid, 45.7 m
(150 ft) per roll; Paper type: Thermal-sensitive;
Chart speeds: 25, 50 mm/sec. +/- 2%
Printout formats: 1-channel automatic or manual Acquisition:
Lead selection: I, II, III, aVR, aVL, aVF, V1, V2, V3, V4, V5, V6;
Modes: Automatic or manual;
Frequency response: 0.05 Hz to 30 Hz, -3 db (filtered)
Meets or exceeds ANSI/AAMI EC11-1982 standard
Input impedance: >50 Mohms; Electrode
Offset tolerance: +/- 300 mV; A/D conversion: 12 bit, 5 uV LSB