Start a Forum
The built in charts are okay when building your dashboards, but you can get even more if you build them from macros. I have built them for a facility view, multi-facility view, and for individual technicians. It is a great way to get a snapshot of how you are performing in the micro as well as the macro. I track open WO, recall status, incidents, PM percentage, hours of productivity, and more. If you want some help, contact me and I can see what I can do for you.
Radiated broadband noise: 10.Hemodialysis Machine (degraded motor brushes) 9. Floor Buffers (degraded motor brushes) 8. Programmable Controllers for Fire Alarm Dampers 7. Degraded HVAC circuit relays - total in 1 hospital=60+ 6. Elevator motors and controllers 5. Hair Dryers (multiples - in patient rooms) 4. Automated Door Openers - multiples 3. Security Radios and Repeaters 2. Lighting Ballasts 1. Lighting Dimmers and dimmer controllers.
This is a 20 year old instrument that appears to be in very good condition. It has a ceramic barium titanate 1 MHZ soundhead which show no pitting, cracks or other damage. All control keys and displays work and there are no cracks or other signs of damage on the housing. When powered up and the soundhead is applied to the patient, it shuts down after 5 to 10 seconds for what appears to be poor ultrasound transmission. This happens with two different gels. On one occasion I did get continuous US with a combination of medium viscosity get and light layer of water. The head was moved rigorously during this test. I have not be able to get the same results in subsequent attempts. I have tried using the head a various different angles without a notable change in results.. According to the manual the soundhead is to be immersed in water prior to use. There is no mention of gel or other ultrasonic wave conduits in the manual. There is no instruction telling the user to remove the soundhead/wand from water for use. My assumption is that the head can be removed from water for use (some older US machines were meant only for use under water) and can be used with gel. Other users of this model of instrument have reported it worked well with standard ultrasonic water based gels. I am looking for suggestion as to what the cause of this issue is beyond the obvious poor transmission and possible solutions. I am going to continue try lightening the water based gel, but this is unlikely to be the cause of a fairly consistent problem. Were a specific gel type or viscosity required you would think it would be mentioned in the manual. Thanks in advance Bruce Raynor firstname.lastname@example.org
- Main input voltage was set to 208 V ac during installation - IPS fuses OK, AC outputs to drivers checked, in specs - Stator power supplies were checked, 5.2 V, +14.92 V, -14.77 V - Safety and door interlocks checked, work properly. Also, emergency stops - When gantry and acq. Computer is turned on: PSD TEST, psd in head 1 and 2 pressed. SYSTEM TESTS press START, start is pressed. INITIALIZATION, remove patient and press ENTER, ENTER is pressed. INITIALIZATION, press START to initialize axes, start is pressed. After few seconds, system ask again, remove patient and press ENTER and so on - With program PANEL I am able to exercise radial, swivel, lateral, bed linear and height, but no rotation. - All FE functionality test pass Most of FE communications test pass, but all tests related to rotation does not pass. I attached some
Troubleshoot a problem while working with a laser device Hello Good time I have a few candela gentle max Pro laser devices that are 2016 and 2017, and after a while, they need to have a password to continue. I will send the device error message pictures to you. Please respond to my email or in the whts app or telegram app below: 00989154159321 Thank you so much Mahmoud Ghadamyari Instagram : Areviktajhiz