How can I go about setting up a Streamway system workbench on this site? I am the Service Manager and it would be great to have a venue to discuss service related issues with potential and existing customers. If you don't know, the Streamway system is a patented direct-to-drain medical fluid waste disposal system that has unlimited capacity for fluid disposal reducing cost and exposure risks.
the diagrams in the operators manual does not specify which tubing goes in the valve upon installation. service manual difficult to read. I am having backflow of reagents, assuming it is caused by misplaced tubing. Can someone share a photo how it looks in the 6 valves? Or describe which tubing should be in the valve?
Hello, We have recently started seeing error messages pop up stating "Replace Sp02 Cable" or "Replace Spo2 Sensor". After some investigating it was determined from Masimo that their new version of Sensors and Cables have an internal chip which tracks usage. After a set amount of time an error message will appear. The error message will vary depending on the model of equipment being used. The sensor/cable will stop working after 72 hours of the message appearing. The reusable sensors are good for 8,760 hours and the Patient Cables are good for 17,520 hours.