Hi, we are a medical charity working in the war zone of Idlib, Syria.
Right now, we are having major issues with a Philips Allura Xper FD10 Cath lab. this is in one of our charity hospital. the power board in the m cabinet keeps resetting during procedures and locks all movements of the table and stand. this is the message we are getting:
geometry is restarting: do NOT change SID
after this, sometimes it starts working again with "limited" movements. but most of the time it totally locks all movements.
Please can you help. RE: Philips - Allura Xper FD10