Hello Guys, i get this error in the device 2d works fine but when i connect two 3d probes in any slot i get this error. I believe the problem is from EPROM in Board Detector Board.Probes works tested If the program is not corrupted i am able to take backup and reprogram the EPROM otherwise if anyone has the Program i will appreciate Thank you in advance for you time
During an injection our unit is pushing fluid from the A-side to the B-side and vise versa while injecting, the unit passes all PM test other than this issue. We are using Med Rad Disposable Sets. I have inspected all gears, belts and functions with no obvious worn parts.
Med Rad Stellant D
Hi Guys! My Sharplan 40C only reached a 0.05 Watts at 38.3mA during Power In and Power Out diagnostics test. What could be the problem? I sent the laser tube for refill in the U.S.(www.parallax-tech.com) is the company that refilled my laser tube but same thing happens when I installed back the newly refilled CO2 laser tube. You can reach me to my e-mail address: firstname.lastname@example.org