Problem Desc: The original problem was that Error 711/80 was displayed when attempting to flouro. Gaskets and corona disks were replaced on HV cables at the tube end, and silicon oil was applied. Error 711/80 was cleared, but images were out of focus/blurry. After continuing to flouro, image quality improved until the problem was resolved. A few days later, out of focus/blurry images were reported with no error. Currently unable to reproduce the problem, system is functioning properly.
What does Error 711/80 mean?
What could cause intermittent out of focus images?
Has anyone ever seen this problem and what was the solution? RE: Siemens - UROSKOP Access