GE Healthcare OEC 9800 Plus Forums
Hi, Have a OEC 9800 Super C 2004. Trying to Project the Image on a External Monitor. I am not a Technician but a Physician. The output on the back of the Workstation is Analog signal (BNC)? How do I "Convert it" to a VGA or HDMI. Saw online that I need a Sewell Luna BNC to VGA/HDMI converter. I was wondering if anyone has done this in the past and can help. What is the Easiest Way to Project the Image (Right & Left) onto an External Monitor. What is the Resolution coming out of OEC 9800? Thanks in Advance.
Hi I have a system that has no errors being reported but while testing I found the KV in some shots( mostly higher and max dose shots, ) are over 6% of kv on c arm and monitor from the reading on my meter. Some are as much as 15% higher on the meter. Ive tried a different meter and get the same results. Ive done filament cals and saved to system twice. The complete generator cal has been done . Ive tried a new high voltage reg pcb