GE Healthcare OEC 9800 Plus Forums
Hello friends. I am having problems with an oec9800 machine. The begining was the Fluoro Function Board that stuck the boot up process remainig only squares in the display of C.Arm. When ia replece this board with another one of different machine it work fine. I purchase a new board but this one still stuck the boot process. Even the workstation is unable to finish loading sofware. I cant connect via RUT because the generator dont finish the boot process. I tryed to erase flash but it didnt work. Somebody can help me? I work with this Carms since 2002 but with the models 9000, 9400 and 9600. Thanks !
Hello my fellow Engineers, I have a 9800 Plus c-arm that has a rotor that constantly spins and the only way to stop it is by unplugging the batteries or disconcerting the tube connector. Please help! I've replaced the batteries and xray tube already.
Hi, Have a OEC 9800 Super C 2004. Trying to Project the Image on a External Monitor. I am not a Technician but a Physician. The output on the back of the Workstation is Analog signal (BNC)? How do I "Convert it" to a VGA or HDMI. Saw online that I need a Sewell Luna BNC to VGA/HDMI converter. I was wondering if anyone has done this in the past and can help. What is the Easiest Way to Project the Image (Right & Left) onto an External Monitor. What is the Resolution coming out of OEC 9800? Thanks in Advance.
Hi I have a system that has no errors being reported but while testing I found the KV in some shots( mostly higher and max dose shots, ) are over 6% of kv on c arm and monitor from the reading on my meter. Some are as much as 15% higher on the meter. Ive tried a different meter and get the same results. Ive done filament cals and saved to system twice. The complete generator cal has been done . Ive tried a new high voltage reg pcb