i have this machine and just of recent it stopped working, it can power on but it does nt complete the initialising process. when it comes to loading the application software it only stops halfway. thus no display of user interface.i would like to replace the software and thus i need a fully loaded harddisk.could anyone send mea quote on firstname.lastname@example.org thank you
We have a Burdick Quest T55 Treadmill system. This morning had CMOS checksum error. Thus time clock battery changed. System then completed booting up. Able to do chemical stress tests but if put in Bruce Protocol reads TREADMILL CABLE NOT DETECTED. Also, have to reboot after each test to enter new test data instead of selecting "New Test." What settings need to be reset. I am also searching our user manual.
Hello all, I have a 7100 that will not ventilate when switching from bag to vent mode. I have checked the microswitch on the lever and the GUI does show indicate which mode its in. I can here the vent engine actuating but I get no movement from the bellows and no movement of air across the flow sensors, thus I get no waveform on the GUI. I have done all leak tests in service mode and in normal operation mode, with no leaks. I have done all calibrations and diagnostic checks in service mode with no problems. The s/w is 1.3. Any help insight would be much appreciated. Thank you