General Comments Posts74 Results
Last Reply Tue Oct 10 2017
I have a Getinge 533HC-E sterilizer that will only pull about 18-20 in vacuum. what could be causing this issue? water pressure is 60PSI, Boiler pressure is 40-50PSI, heating elements are good and receiving proper power, the jacket holds 30PSI, the chamber reaches 30-32PSI. What should I check next?
Thank you for your kind reply. Please find below more information about further troubleshooting by our technician: Here is the findings of our technician just re-visiting the site to do further troubleshooting as advise. - I did the reset and it was still the same thing - So I did a test on the RU - The error on the RU is 11700 - The result on the test 13720 points to the fan in the manual - I open the place of the fan its missing (we can perhaps source this locally) Please advise and give us quote for the replacement should this be the conclusion. Thanks.
Last Reply Fri Sep 29 2017
I have with Capsula X, though it erazed well but with a beeping sound error of erasure lamp fault, though fuses were checked to be o.k. Should we then replace the entire Erasure unit to remove the sound and error, though still erasing, but the customer is not happy with the error and sound. Kindly give you advice and part number for the this part and dealer contact. Thanks. Gafar i did a secondary erasure and i see lamps ??? i am assuming it is lamps but do not know how to replace them .
Last Reply Tue Mar 21 2017
Hospira has been co-selected for the AAMI Foundation/Institute for Technology in Health Care Clinical Award. Its Plum A+ infusion pump and Mednet safety software system have helped Lancaster General Hospital achieve error-free medication delivery and improved patient safety. See the full news story here: http://www.aiv-inc.com/iv-pump-news/hospira-and-baxter-sigma-infusion-pump-systems-receive-awards.html