I have a Cardiocap/5 that had the gasses calibrated in 8/2015 - now the O2 is steady at 21% (atmosphere). Tried recalibration , would allow me to calibrate all gasses except O2.
Have a planned maintenance kit on the way, just wondering if there was any other possible correction that needs to be done
I am wondering if you resolved this issue?
I am currently working with one with the same issue. Possible tubing leak?
If you are in need of immediate assistance please feel free to reach out directly to our Telemetry-Central Station sponsors. They are happy to help out when they can.
Just be sure to tell them you were referred by MedWrench.
The O2 cell is bad and needs replaced. We can repair this for you.