MedWrench is available to help people with specific questions and troubleshooting problems. Could you please be more specific with the issue that you are having with your instrument, and the MedWrench team will do their best to assist you in solving the issue. This will also help other community members who face similar challenges.
Mia DeBiase Sales & Marketing Assistant email@example.com
This is actually a FUJI Reader but if you et a network communication error the Reader can not talk to the Workstation Console. Check switches and network cables
OK. we have Philips PCR-Eleva s hi-res, model - CR-IR 368
The issue is when we instell PCR softwear,there is network commnication problem and the CR is displays OFFLINE.
shear your solution with me pls
take a stand alone switch and plug reader and workstation into switch making sure IP are on the same subnet and the install PCR Software, thus taking it off or the Whole network or making it stand alone while loading software
I appreciate your fast response Frank, hope it will be helpful;
thank you again