flow cell adjust
you will require a service manual and some special tools for this assignment ,do you have these things ?
yes i have the service manual and the latex particles. i thought i had it alighned good, the CV's are where they need to be. controls pass. but the patient samples are all running a little high on the Baso. i did replace the laser. could that be the difereance from a week laser to a strong laser. Thank you.
The Baso can not be higher regardless of having changed the laser tube .
make a fresh latex ( latron ) and fine tune the CV's to close o perfect
Did you change the laser and laser Power supply unit ? , some times it is better to change the power supply as well because a new tube could wear out the older laser power supply .
With good alighment
I feel that you are nearly there .
Are you using fresh blood samples for patient samples , from how many hours ago ? Are the baso results high on all the patient samples ? If possible draw blood and run a fresh whole blood sample .
baso raise as whole blood in the tube gets older .
i would lik to send you pictures of the CV's i got, patient sample, and controls. to look at the histogram and tell me what you think. i could email or text them. i dont think there is a way to do it on this site.
The following items are necessary:
woc fine procedure.
Excel spreadsheet to find new gains.
laser power meter.
latex of 7 um.
laser protection glasses
fresh blood from patient with less than 04 hours.
the information and procedures I can send it to you by mail