MedWrench is available to help people with specific questions and troubleshooting problems. Could you please be more specific with the issue that you are having with your instrument? The MedWrench team will do their best to assist you in solving the issue. This will also help other community members who face similar challenges.
Help ... Manual service DRS 601
Alexandra Wright MedWrench - Sales & Marketing Coordinator email me: email@example.com
It is section 8-7 in the service manual. For 220/240V operation, remove the short on the right-hand end of the terminal block (on the 24V switching regulator).
[(Note to administrator: I have a copy I can send in. Tell me how.]