I set the cutting to "60" on the display and when I cut it will flash a "P" on the right side of the number. Can anyone tell me why a "P" comes on when cutting ?
Tue Mar 24 2020
Cooper Surgical Leep System 6000
Tue Mar 24 2020Reply from Chris S
This is what the service manual states:
CONDITION - No Monopolar:- RF Flashing "P"
POSSIBLE PROBLEM - Internal error
REMEDY - Monopolar RF generator, B+ power supply, or controller
Official MedWrench Guru