Arcadis Gen2 Errors
MSY_4191: X-Ray aborted.
BIC_10105: Is related to the CIPP/DVP boards in PC.
Thinking simplisticly, I would suggest checking the X10 connection (the white square connector on generator.
Check to make sure it was insered correctly.
Check all the male pins are OK, (not bent), a couple of pins are slightly shorter, its normal.
I did not notice any bent pins. I disconnected and connected the umbilical a couple of times to verify it was connected fully and correctly. At almost a full minute of fluoro the ASP_BIC_E_10105 comes up on the workstation. I also noticed Err 4500 on the C.
I assume we are possibly now looking at a PC issue?
10105 & 4500 error are general image system messages. Any interuption or connection issue can cause these issues.
The Coax (Gigga link) cable from the BNC CIPP (PC end) connection up through the SG cable (X10) to the TV Camera can interfere with data transfer & cause the PC to be overloaded with error mesages, then 10105, 4500 etc. etc appear.
You could mesure the resistance from the TV Camera BNC to the CIPP BNC, while measuring move/twist the SG & main C-Arm cable/s to see if resistance changes, do same with BNC shield. This cable, by the way is also a TRIAX so don't be confused measuring the wrong ground (BNC)!
Another thing is to make sure all the Ground cables in TV camera are correctly clamped.
Make sure the coax cable in the TV camera Hood are not bent sharply, they must be a smooth gentle bend.
Make sure ground/s inside the generator where the D1 & BNC Coax/triax connections are.
Also, you may try cleaning the CIPP & DVP PCI connections within the PC (Be careful not to damage any surface mounted components).
One final thing, I cannot stress this enough, check the coax cable within the X10, if its kinked/sharply bent this can also give these errors as the cable impedence is compromised!
If you have a gen2 system, check the additional X11 cable connection where the SG cable exits the monitir cart base. The latest cables now no longer use this X11 as it also casued problems.
There was also a modification when an additional bracket is used where the cable comes out of SG cart.
These documents are available on the Siemens Healthineers document website.
By the way, I would not replace the PC (yet). As I recall about 80% of PC's returned are not faulty.
The vast majority suffer BIOS battery failures & image system connection issues.
I have a problem with a 2nd generation Siemens Arcadis Varic. Both screens work and are controlled. The equipment cart (C-arm not connected) starts the Windows normally, then the Siemens software and the "counter" of the Syngo software runs. After about 30 there will be the error message ASP_ASM_E_70 or sometimes ASP_ABE_E_222. Has anyone ever fixed such a bug? bitlife pc jiofi.local.html
Error 70 is a memory related error.
This memory could be releated to the Grahic card configuration for monitors, the graphic card config should be 'Dual' mod, nothing else.
After the errors appear what happens, does syngo continue counting? Does the count rolgo 99 & then tollover back to one??? Or does it stop?
Can you get into Explorer?
What model PC do you have, M450, M460, M470, M720 etc.
When the system starts, you will see screens like BIOS do you see the 3 HDD listed?
There is a chance the BIOS had had a bad day, can you check the BIOS. Do not reset BIOS unless you have the crrect BIOS CD/Model dependent.
Can you give more details, how old is system, Varic/Orbic what SN range IE: Varic 12*** Orbic 22***, just provide the third number of SN like Varic123**, Orbic 223**.