Need GUI software & instructions for voltage adjustments
Just to be clear, your saying this new Spellman II power supply does not have any dip switchs or pot adjustments?
Siemens did change the II power supply model where the adjustment voltages remained the same but the voltage ratios to be adjusted changed & therefore you had to make corrections based upon the original II test certificate voltages hen compared to the new II power supply adjustment ratios.
Yes, it has a D15 connector for adjusting G1, 2, 3 using a software GUI. The plate voltage is adjusted using a pot. on the IIPS. Only 3 wires connect the IIPS to the c-arm (power, ground, and Mag) so you disconnect the D15 connector (under the camera cover) to attach to the IIPS, and make adjustments there.