You can adjust your Community Subscriptions in Settings
You can add Community Subscriptions in the search bar that says "Subscribe to more communities ..."
Equipment: Syneron-Candela - CO2RE
First question: Are the aiming and treatment beams coaxial?
Second question: If they are, does the aiming beam stay normal when the arm is moved around? ("Normal" meaning not diffracted.)
They are not. The laser beam is aligned, while the aiming beam is not.
With the arm off, perform a near/far alignment of the aiming beam conforming it to the treatment beam, and you'll be good to go.
© 2017 MedWrench, LLC. All Rights Reserved