Handpiece won't go into cooling
Hello Laser J,
Are you actually getting any kind of error code when you try to enter into cooling? Are there any indicators your system is giving at all?
Sounds like you have a TEC (Thermo Electric Cooler) issue. They go bad over time and could require replacement. Another possibility is that the heatsink compound is old and allow for good heat tranfer. But either way, I believe the issue to be TEC related.
Senior Laser Engineer
Would you know where that is located on a Xeo? Is it an entire TEC circuit that needs to be replaced, or just a single component/ part?
Excuse me, I am a little new to this industry, and not totally famiiliar with the laser. The error is not with the handpiece nor with the YAG. I do not know what you are refering to as the "crystal light guide". Is the TEC on a board? If so which one. If not, where in the laser could I find this TEC.
Thanks for your help