Equipment: Nikkiso - DBB-07
This indicates that dialysate pressure has exceeded whatever the preset level to trigger this alarm is.
- Is the UF volume too high?
- Is the dialser's UF factor too small?
- Is there air present on the dialysate side? If so, press "Bypass select" key and then "Gaspurge." After the air has been removed, press "Gaspurge" again, "Bypass select" again, and "Alarm reset" in order to continue treatment.