You can adjust your Community Subscriptions in Settings
You can add Community Subscriptions in the search bar that says "Subscribe to more communities ..."
Actuator board failed to send data
Equipment: Fresenius 2008T
IF “Failed Sending Data To Actuator Board” AND a “24V” alarm has NEVER occurred:
Replace the Actuator board
© 2017 MedWrench, LLC. All Rights Reserved