You can adjust your Community Subscriptions in Settings
You can add Community Subscriptions in the search bar that says "Subscribe to more communities ..."
Occasional C05 (unstable signal) when starting BP. Issue is not related to cuff. Thank in advance.
Equipment: welch Allyn spot vital sign
Do an NIBP calibration with the Welch Allyn Service tool.
© 2017 MedWrench, LLC. All Rights Reserved